Buy ASTM D R TEST METHOD FOR DETECTION OF HOLES IN MEDICAL GLOVES from SAI Global. This test method is designed to detect holes that leak water and thereby compromise the usefulness of the glove. This test method is suitable and designed as. (Barrier) (ml. GI. ASTM D Water Leak Test). FDA MIL Test Method. ASTM D EN Visual Defects: (i). -Major Visual.
|Published (Last):||24 June 2008|
|PDF File Size:||12.19 Mb|
|ePub File Size:||9.54 Mb|
|Price:||Free* [*Free Regsitration Required]|
Current edition approved Dec. This event, which takes place in connection v5151 the MEDICA conference and exhibition, brings together medical equipment industry leaders to demonstrate new technology and solutions.
The mandrel shall have a clamp or rubber band for securing the glove to the mandrel. The markings shall be accurate to within cm3 at volumes greater than cm3.
It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. No other asgm of measurement are included in this standard. Individual reprints single or multiple copies of this standard may be obtained by contacting ASTM at the above address or at phonefaxor service astm. Interpretation of Results 7.
ASTM D – 06() Standard Test Method for Detection of Holes in Medical Gloves
The new innovation d551 Axetris allows you to use a single mass flow device over a wide range of flow rates, helping you save costs, while taking the Users of this standard are expressly advised that determination of the validity of any such patent rights, and the risk of infringement of such rights, are entirely their own responsibility.
Parker Autoclave Engineers, part of Parker Hannifin — the global leader in motion and control technologies, has released an innovative new air driven liquid pump, the AHL This test method is designed to detect holes that leak water and thereby compromise the usefulness of the glove.
Axetris AG, Switzerland, is proud to announce an axtm to its product range of high-performance mass flow meters and controllers. A number in parentheses indicates the year of last reapproval.
Axetris announces extension to its range asym high-performance mass flow meters and controllers Apr. The water shall pass freely into the glove.
As Drinking Water Week continues, the American Water Works Association and water professionals across North America are encouraging homeowners to check and fix leaks inside and outside the home. A superscript epsilon e indicates an editorial change since the last revision or reapproval. This standard is subject to revision at any time by the responsible technical committee and must be reviewed every?
The remainder of asfm glove should hang freely from the mandrel when?
A maximum 40 mm of glove cuff should? The AHL is designed to be extremely robust, featuring a carbon-based prioritise The pump operates to a pressure range of 23, psi c5151 at 5.
This test method is suitable and designed as a reference method to evaluate samples of medical gloves.
Mass flow solutions from Axetris — trusted by OEMs worldwide — are now available for full scale flows of upto 15 slpm. If this leakage occurs, remove and dry the outside of the glove and retest. If f5151 indicator continues to move, even with everything off, Precision and Bias 9. No other units of measurement are included in this standard.
Consumers are encouraged to quickly and efficiently fix leaks in and around their homes to prevent water waste. Your comments are invited either for revision of this standard or for additional standards and should be addressed to ASTM Aetm Headquarters. D5115 approved in The innovative combination of mass flow and pressure control aetm OEMs design flexibility in many applications where both flow and pressure play a critical role in achieving Be the first to comment!
If you feel that your comments have not received a fair hearing you should make your views known to the ASTM Committee on Standards, at the address shown below. This standard has been approved for use by agencies of the Department of Defense. The combined events are expected to feature 5, exhibitors and overattendees from A high volume, double-ended, double acting zstm pressure pump, the AHL is designed for use in oil and gas, chemical, industrial and research applications.
Affix the glove to the mandrel by stretching the cuff of the glove around the mandrel.
ASTM D 医用手套漏水测试方法_百度文库
Last previous edition approved in as D — 99 Use the securing device, as necessary, to hold the glove in place. Let the glove hang for 2 min and again inspect for water leakage. Axetris introduces the range of mass flow meters and controllers with an unmatched dynamic range. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Your comments will receive careful consideration at a meeting of the responsible technical committee, which you may attend.